BRAND offers the ideal pipette for every hand: BRAND’s new Transferpette® S. Transferpette® S models are are the perfect manual pipettes for demand-. Transferpette® electronic. F I R S T C L A S S · B R A N D. Gebrauchsanleitung. Operating Manual. Mode d’emploi. Instrucciones de manejo. 3. -8/ . Testing Instructions (SOP) testing of the piston operated pipette Transferpette®. BRAND additionally provides a calibration service (see chapter 7).
|Published (Last):||8 January 2016|
|PDF File Size:||10.4 Mb|
|ePub File Size:||7.62 Mb|
|Price:||Free* [*Free Regsitration Required]|
Inspect the pipette tip cone for damage. Pipetting c Let the pipetting button slide back slowly. Set nominal volume Set volume to the maximum volume indicated on the instrument see page 32 for procedure.
Inspect the piston and O-Ring-seal for contamination. The cycle can, however, be adjusted to individual requirements.
The form can also be downloaded from www. Safety Instructions Safety Instructions Please read the following carefully!
Don’t have an account? Table of Contents Add to my manuals Add.
Optimum analysis results can only be obtained with quality tips. Inspect the piston and seal for contamination. Consult the manufacturer for use in temperatures outside of this range. If the pipette operation is clearly inaccurate, or if the instrument must be adjusted for solutions of different densities and viscosities or specially- shaped pipette tips, adjustments can be made using the Easy Calibration Technique.
Eject the pipette tip. Page 17 Inspect the pipette tip cone for damage. It is beyond the scope of this manual to address all of the potential safety risks associated with its use in such applications. Safety Instructions Please read the following carefully!
BRAND Transferpette S Operating Manual
Page 9 Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop. Parts will differ slightly depending on nominal volume of instrument. Also See for Transferpette S Operating manual – pages. Limitations Of Use Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity.
Laboratory Equipment Brand dispensette Operating Manual pages. Adjustment — Easy Calibration The instrument is permanently adjusted manuwl aqueous solutions. Page 8 Pipetting c Let the pipetting button slide back slowly. Pipetting Pipetting 5 ml and 10 ml instruments should transferpettee be used with the PE filter installed see page Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop. Just send in the instruments to be calibrated, accompanied by an indication of which kind of calibration you wish.
Don’t lay the instrument horizontal when the tip is filled. Limitations of Use The instrument transfwrpette intended for the pipetting of liquids within the following limitations: Ensure that the tip is securely seated.
BRAND TRANSFERPETTE S OPERATING MANUAL Pdf Download.
Comments to this Manuals Your Name. Discharge sample a Place the pipette tip against the wall of the vessel. This instrument may sometimes be used with hazardous materials, opera- tions, and equipment. For further information, refer to the accuracy table on pages 37 and To print the manual completely, please, download it.
It is the responsibility of the user of this pipette to consult and establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. In order for the liquid to reach its end position, leave the tip immersed for a few seconds. Checking the Volume Depending on use, we recommend inspection of the instrument every 3 to 12 months. Pipette tips are disposables items!
Enter text from picture: To do this, affix a tip, and aspirate a sample. Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page