The main task of technical committees is to prepare International Standards. Draft International Standards adopted by the technical committees are circulated to. à aiguille interne. STANDARD. ISO. Second edition. Reference number. ISO (E). Provläsningsexemplar / Preview. STANDARD. ISO. Second edition. Reference number. ISO (E). This is a free 4 page sample. Access the full version online.
|Published (Last):||12 March 2018|
|PDF File Size:||4.23 Mb|
|ePub File Size:||19.7 Mb|
|Price:||Free* [*Free Regsitration Required]|
EN ISO – European Standards
I Principle Water is allowed to flow through the catheter and the amount of flow is measured either isso or gravimetrically. Available from Munsell Color, N. An approved test method for producing a value of radio-detectability will be established.
Certain conventions are, however, not identical to those used in Indian Standards. For the purpose of deciding whether a particular requirement of this standard is complied with, the final value, observed or calculated expressing the result of a test or analysis, shall be rounded off in accordance with IS 2: Pages 1 and 2 4. Attention is particularly drawn to the following: E-5 Test report The test report shall contain at least the following information: In this adopted standard, reference appears to certain International Standards for which Indian Standards 105555-5 exist.
Review of Indian Standards Amendments are issued to standards as the need arises on the basis of comments.
Details of which is given in National Annex A. Blood flow into the needle hub. I Principle A force is applied successively tensile and connpress- ive to the needle tube and needle hub and the tube- hub union is then examined for loosening. Limited collection of colours. Fitting attached to the needle tube, providing communication with its bore. Round the calculated average water fiowrate to the nearest whole oso of millilitres. Page 3, subclause 4. Standards are also reviewed periodically; a standard alongwith amendments is reaffirmed when such review indicates that no changes are needed; if the review indicates that changes are needed, it is taken up for revision.
Intravascular catheters ISO , -3, -4, -5
The fluid is allowed to flow into the needle, and the time taken for fluid to leak through the vent fitting is measured. Measure the time taken for fluid to form the first falling drop at the back of the vent fitting. The details of conditions under which the licence for the use of the Standard Mark may be granted to manufacturers or producers may be obtained from the Bureau of Indian Standards.
The fluid pathway in the needle shall be free of unintended obstructions that would prevent flashback.
If a steel tube is used, it shall connply with ISO Add the following reference: Other parts of this standard are as follows: All standards are subject to revision, and parties to agreements based on this part of ISO are encouraged to investigate the possibility of applying the 1055-55 recent editions of the standards indicated below.
Bureau of Indian Standards BIS is a statutory institution established under the Bureau of Indian Standards Act, to promote harmonious development of the activities of standardization, marking and quality certification of goods and attending to connected matters in the country.
The number of significant places retained in the rounded off value should be the same as that of the specified value in this standard.
BS EN ISO 10555-5:2013
This Indian Standard has been developed from Doc No.: Figure 1 — Typical over-needle peripheral intravascular catheter NOTE 2 At the time of publication of this part of ISOthere is no acceptable, validated test method to determine radio-detectability. Catheter designed for the introduction or withdrawal of liquids or devices into isl from the peripheral vascular system. This does not preclude the free use, in the course of implementing the standard, of necessary details, such as symbols and sizes, type or grade designations.
The catheter tube may have a single lumen or multiple lumens. Fixed or removable fitting per- mitting venting of air while restricting or preferably preventing the escape of blood.
CSN EN ISO 10555-5
When tested in accordance with annex E, fluid shall not leak out of the vent fitting within 15 s. No part of the these publications may be reproduced in any form without the prior permission in writing of BIS. Annex D shows examples of typical needle point geom- etries.
NOTE To improve the visibility of the solution, a colorant such as red or blue food dye may be incorporated.